- Open Access
A new resorbable device for ligation of blood vessels - A pilot study
© Höglund et al; licensee BioMed Central Ltd. 2011
- Received: 23 March 2011
- Accepted: 8 July 2011
- Published: 8 July 2011
During surgery, controlled haemostasis to prevent blood loss is vital for a successful outcome. It can be difficult to ligate vessels located deep in the abdomen. A device that is easy to use and enables secure ligatures could be beneficial. Cable ties made of nylon have been used for ligation but the non-resorbable material caused tissue reactions. The objective of this study was to use a resorbable material to construct a device with a self-locking mechanism and to test its mechanical strength and ligation efficiency.
The device was manufactured by injection moulding of polydioxanone, a resorbable polymer used for suture materials. Polydioxanone with inherent viscosities of 1.9 dL/g and 1.3 dL/g were tested. The device consisted of a perforated flexible band which could be pulled through a case with a locking mechanism. After a first version of the device had been tested, some improvements were made. The locking case was downsized, corners were rounded off, the band was made thicker and the mould was redesigned to produce longer devices. Tensile tests were performed with the second version.
The first version of the device was used to ligate the ovarian pedicle in a euthanized dog and to test echogenicity of the device with ultrasound. Compression of vessels of the ovarian pedicle was examined by histology. Both versions of the device were tested for haemostasis of and tissue grip on renal arteries in six anaesthetised pigs.
The tensile strength of the flexible band of the devices with inherent viscosity of 1.9 dL/g was 50.1 ± 5.5 N (range 35.2-62.9 N, n = 11) and the devices with inherent viscosity of 1.3 dL/g had a tensile strength of 39.8 ± 8.1 N (range 18.6-54.2 N, n = 11). Injection moulding of the polymer with lower inherent viscosity resulted in a longer flow distance.
Both versions of the device had an effective tissue grip and complete haemostasis of renal arteries was verified. The device attached to the ovarian pedicle could be seen with ultrasound, and vessel compression and occlusion were verified by histology.
Tests of functionality of the device showed complete haemostasis and good tissue grip. Devices with a band of sufficient length were easily applied and tightened in tissue.
- Renal Artery
- Injection Moulding
- Inherent Viscosity
- Flow Distance
- Complete Haemostasis
Bleeding associated with surgery is a potentially serious complication. Haemostasis to prevent blood loss is commonly achieved by tying a surgical thread around the blood vessel. Vessels localized deep in the abdomen can be difficult to ligate . A device that enables secure ligatures that are easily applied could be beneficial.
The use of non-resorbable nylon cable ties for haemostasis in soft tissue surgery has been described [2–8]. However, non-resorbable materials used for ligation purposes and left permanently in tissue, can cause complications such as fistulas and granulomas [9–11]. The advantages of cable ties for ligation purposes were ease of application, quick ligation procedures and good "knot" security [6–8]. If a device, based on the construction of a self-locking loop, could be made of a resorbable material it could minimize the risk of adverse tissue reactions but potentially maintain the advantages experienced with traditional cable ties.
The aim of this study was to develop a device of a resorbable material, which enables reliable and easy ligation of blood vessels.
Design of the device
Design of moulds for injection moulding
Injection moulding of polydioxanone
Polydioxanone (Resomer® X, Boehringer Ingelheim Pharma GmbH, D-55216 Ingelheim, Germany) was chosen for the material of the device, a synthetic resorbable polymer which is widely used for suture materials and implants since many years. It is flexible and elastic and its induction of inflammatory reactions is minimal. It is degraded through hydrolysis and may be completely resorbed within 6-12 months [12, 13].
The polymer was heated above melting temperature and injected into the mould using an in-house built injection moulding machine. High pressure was maintained after injection and crystallization was allowed to occur for about one minute before the mould was opened. Polydioxanone of inherent viscosity of 1.9 dL/g was used for injection moulding in the first version's mould. Two batches of polydioxanone with inherent viscosities of 1.9 and 1.3 dL/g respectively were tested for injection moulding in the second version's mould.
The tensile strength of the flexible band was measured using a 5544 Single Column Testing System (Instron, USA). Samples were prepared by cutting away the locking case of the device and the solid part of the flexible band, leaving approximately 4 cm of the perforated band. The samples were clamped 1 cm from each end and pulled to break at a rate of deformation of 40 mm/min. The tensile strength was determined as the maximum failure load. Eleven samples manufactured from each of the polymer batches of polydioxanone were tested. All tested devices were of the second version. The level of statistical significance was defined as p < 0.05 (Student's t-test, equal variances not assumed).
One dog weighing 20 kg, which was euthanized prior to the test due to reasons not associated with this study, and six pigs weighing 25-28 kg, which were anaesthetised for other reasons at Uppsala University Hospital, Sweden, were used in the study. The Uppsala Animal Ethics Committee, Sweden, approved this additional test (permission number C172/8).
Test of the devices in tissue
Ovarian pedicle, imaging and histology
An incision was made from approximately 2 cm cranial of the umbilicus along the midline of the abdomen in the dog. The linea alba was incised, the abdomen opened and the uterine horns and ovaries were localized. A hole was made in the broad ligament close to the left ovary. A loop was formed around the ovarian pedicle and the band was pushed into the locking mechanism (first version of the device). The loop was tightened, compressing the tissue inside the loop. The surplus band extending beyond the channel exit in the locking case was cut off.
The echogenicity of the device was examined with ultrasound (Sequoia 512, Acuson, Siemens AG Medical Solutions, Germany) with an 8.0 MHz linear probe. The probe was placed on the abdomen after application of ultrasound gel on the skin. After the investigation the device with adjacent tissue was removed and embedded in resin (Historesin, Leica Microsystems Nussloch GmbH, Germany). After dehydration in increasing concentrations of ethanol, followed by increasing concentrations of water-soluble resin, the sample was embedded in 100% resin. Sections were cut using a microtome (Leica RM 2165, Leica Instruments GmbH, Germany) with glass knives as close as possible to the device. The samples were mounted on glass and stained with hematoxylin and eosin (HE).
Test of haemostasis and tissue grip of renal arteries in six pigs
The abdomen was opened midway along the linea alba and both renal arteries were free-dissected in two (first version) and four pigs (second version), respectively. A loop was formed around the artery with the device and the loop was tightened causing compression of the vessel. When the band was fully pulled through the locking case of the device, the artery was cut between the kidney and the device about 0.5 cm from the device. The renal artery with the attached device was inspected for haemostasis for about 5 minutes. The ability of the device to withstand a ligature slip-off was tested by applying a force of 10 N using a dynamometer attached to the device.
Arterial blood pressure was registered by invasive continuous measurements in the femoral artery of the pigs and was recorded once a minute (SC 9000XL, Siemens Medical Solutions, USA). The least square means of systolic and diastolic blood pressures during ligation were calculated.
Injection moulding of polydioxanone
The mould for version one resulted in short devices whereas the modified design of mould for version two improved manufacturing results. The polymer with an inherent viscosity of 1.9 dL/g resulted in a shorter flow distance in the moulds and made the flexible band too short for optimal handling during surgery. The polymer with an inherent viscosity of 1.3 dL/g resulted in a longer flow distance, 10 cm, and the second version's mould was filled. A higher temperature of the melted polymer and mould facilitated flow distance in the mould but a higher temperature of the mould increased the time needed for crystallization. At least one minute was necessary for crystallisation of the polymer to occur. The material of the device expanded if sufficient crystallization had not occurred when the mould was opened, which caused constriction of the channel through the locking case.
Tests of first and second version of the device in tissue
The production of the resorbable device was successful and the tests of functionality in animal tissues showed that an effective tissue grip and complete haemostasis were achieved.
The use of polydioxanone of higher inherent viscosity (1.9 dL/g) increased the tensile strength of the device but was a limitation in the manufacturing process due to shorter flow distance. The flow distance can be increased with higher temperature of the mould, but then the time needed for crystallization may have to be shortened by cooling the mould. The optimal process temperature during injection moulding is dependent of desired flow distance, diameter of the mould's channels, inherent viscosity of the polymer, its melting point and injection pressure. A higher temperature during the manufacturing process and lower inherent viscosity (1.3 dL/g) facilitates flow but temperature has to be balanced against detrimental change of properties of the polymer . The inherent viscosity of the polymer also affected mechanical characteristics of the device, as would be expected by the difference in molecular weight. The results of the tensile test are similar to other studies of suture material (polydioxanone 3-0) at time zero [15, 16] but comparisons should be done cautiously as the methods used are not identical.
Both versions of the device were tested on free-dissected renal arteries. Ligatures on the renal artery in connection with nephrectomy must be strong enough to prevent bleeding due to the high blood pressure in these vessels. Two ligatures  or two clips on the renal artery [18, 19] are therefore regularly applied for security reasons. The device, however, has a built-in double security and yields two ligations similar to a double ligature. The renal arteries in the pigs, pressed towards the protrusion and the ridges, were squeezed into a zigzag pattern which created haemostasis. A double ligation is achieved by pressing the artery against the two ridges, one on each side of the locking case. If one side of the band breaks the flexible band cannot be reversed through the locking case. The construction also prevents the device to slide off the tissue. The renal arteries of the pigs were only observed for a short time, but complete haemostasis was obtained. Sufficient length of the device is important for ease of application in tissue and longer devices were successfully manufactured with the mould of the second version of the device and a lower inherent viscosity. A smooth and easy passage of the band through the locking case with a trustworthy mechanical performance of the locking mechanism of the device is essential for safety.
In the dog, the device effectively compressed the vessels together with the surrounding fatty tissue in the ovarian pedicle as verified by histology. The size of the self-locking loop of the device suggests that it may be used for ovarian pedicles regardless of breed size. The device could be seen on ultrasound imaging. The polymer may in the future be blended with a radiopaque substance to facilitate x-ray imaging . There were some study limitations. The degradation effect on the polymer during the manufacturing process was not investigated and the tensile test was restricted to the perforated flexible band of a limited number of devices. As tests under hydrolytic conditions over time were not performed, conclusions are limited to initial strength.
A novel injection moulded device made of the resorbable polymer polydioxanone was constructed. Tests of functionality of the device showed complete haemostasis and good tissue grip. Devices with a band of sufficient length were easily applied and tightened in tissue.
The authors thank Agneta Ronéus and Göran Hedenstierna at the Animal Laboratory at Clinical Physiology, Department of Medical Sciences, Uppsala University, Uppsala, Sweden for invaluable assistance and access to the laboratory. We also thank Lena Holm at the Department of Anatomy, Physiology and Biochemistry, SLU, Uppsala, Sweden for tissue preparation enabling histologic evaluation. This project was supported by S Johansson foundation, A Personnes foundation and Almi Företagspartner Uppsala AB, Sweden.
- Berzon JL: Complications of elective ovariohysterectomies in the dog and cat at a teaching institution: clinical review of 853 cases. Vet Surg. 1979, 8: 89-91. 10.1111/j.1532-950X.1979.tb00615.x.View ArticleGoogle Scholar
- McDougall EM, Clayman RV, Chandhoke PS, Kerbl K, Stone AM, Wick MR, Hicks M, Figenshau RS: Laparoscopic partial nephrectomy in the pig model. J Urol. 1993, 149: 1633-6.PubMedGoogle Scholar
- Cadeddu JA, Corwin TS, Traxer O, Collick C, Saboorian HH, Pearle MS: Hemostatic laparoscopic partial nephrectomy: cable-tie compression. Urology. 2001, 57: 562-6. 10.1016/S0090-4295(00)01009-8.View ArticlePubMedGoogle Scholar
- Cokelaere SM, Martens AM, Wiemer P: Laparoscopic ovariectomy in mares using a polyamide tie-rap. Vet Surg. 2005, 34: 651-6. 10.1111/j.1532-950X.2005.00101.x.View ArticlePubMedGoogle Scholar
- Adams RJ, Narayan O: Partial splenic resection using nylon cable tie ligatures in rhesus monkeys. Lab Anim Sci. 1993, 43: 267-9.PubMedGoogle Scholar
- Carpenter R: Nylon bands used as ligatures and fixation devices in small animal surgery. Proc 40th Ann Meet Am Anim Hosp Assoc. 1973, 718-721.Google Scholar
- Zagraniski MJ: Splenectomy using nylon cable tie bands. Feline Practice. 1979, 9: 33-35.Google Scholar
- Zagraniski MJ: Ovariohysterectomy in the pregnant cat utilizing a nylon cable tie band. Feline Practice. 1980, 10: 41-44.Google Scholar
- Werner RE, Straughan AJ, Vezin D: Nylon cable band reactions in ovariohysterectomized bitches. J Am Vet Med Assoc. 1992, 200: 64-66.PubMedGoogle Scholar
- Fossum T: Small Animal Surgery. 2007, Mosby Elsevier, 61-3Google Scholar
- Spackman C, Caywood D, Johnston G, Feeney D: Granulomas of the uterine and ovarian stumps: A case report. J Am Anim Hosp Assoc. 1984, 20: 449-453.Google Scholar
- Middleton JC, Tipton AJ: Synthetic biodegradable polymers as medical devices. Medical Plastics and Biomaterials Magazine. 1998Google Scholar
- Molea G, Schonauer F, Bifulco G, D'Angelo D: Comparative study on biocompatibility and absorption times of three absorbable monofilament suture materials (Polydioxanone, Poliglecaprone 25, Glycomer 631). Br J Plast Surg. 2000, 53: 137-41. 10.1054/bjps.1999.3247.View ArticlePubMedGoogle Scholar
- von Oepen R, Michaeli W: Injection moulding of biodegradable implants. Clinical Materials. 1992, 10: 21-8. 10.1016/0267-6605(92)90080-D.View ArticlePubMedGoogle Scholar
- Greenberg CB, Davidson EB, Bellmer DD, Morton RJ, Payton ME: Evaluation of the tensile strengths of four monofilament absorbable suture materials after immersion in canine urine with or without bacteria. Am J Vet Res. 2004, 65: 847-53. 10.2460/ajvr.2004.65.847.View ArticlePubMedGoogle Scholar
- Nichols S, Anderson DE: Breaking strength and elasticity of synthetic absorbable suture materials incubated in phosphate-buffered saline solution, milk, and milk contaminated with Streptococcus agalactiae. Am J Vet Res. 2007, 68: 441-5. 10.2460/ajvr.68.4.441.View ArticlePubMedGoogle Scholar
- Rawlings C, Bjorling D, Christie B: Kidneys. Textbook of small animal surgery. Edited by: Slatter D. 2003, Philadelpha: Saunders, 2: 1608-1609. 3Google Scholar
- Modi P, Rizvi SJ, Gupta R: Use of Hem-o-lok clips for vascular control during retroperitoneoscopic nephrectomy in children. J Endourol. 2009, 23:Google Scholar
- Ponsky L, Cherullo E, Moinzadeh A, Desai M, Kaouk J, Haber GP, Chen D, Ng C, Fuchs G, Singh D, Finelli A, Frank I, Matin S: The Hem-o-lok clip is safe for laparoscopic nephrectomy: a multi-institutional review. Urology. 2008, 71: 593-6. 10.1016/j.urology.2007.11.015.View ArticlePubMedGoogle Scholar
- Mottu F, Rüfenacht DA, Doelker E: Radiopaque polymeric materials for medical applications: Current aspects of biomaterial research. Invest Radiol. 1999, 34: 323-335. 10.1097/00004424-199905000-00001.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.