- Open Access
Measurement of single kidney glomerular filtration rate in dogs using dynamic contrast-enhanced magnetic resonance imaging and the Rutland-Patlak plot technique
Acta Veterinaria Scandinavica volume 60, Article number: 72 (2018)
Nephropathies are among the most common diseases in dogs. Regular examination of the kidney function plays an important role for an adequate treatment scheme. The determination of the glomerular filtration rate (GFR) is seen as the gold standard in assessing the kidney status. Most of the tests have the disadvantage that only the complete glomerular filtration rate of both kidneys can be assessed and not the single kidney glomerular filtration rate. Imaging examination techniques like dynamic contrast-enhanced magnetic resonance imaging have the potential to evaluate the single kidney GFR. There are studies in human medicine describing the determination of the single kidney GFR using this technique. To our knowledge there are no such studies for dogs.
An exponential fit was found to describe the functional interrelation between signal intensity and contrast medium concentrations. The changes of contrast medium concentrations during the contrast medium bolus propagation were calculated. The extreme values of contrast medium concentrations in the kidneys were reached at nearly the same time in every individual dog (1st maximum aorta 8.5 s, 1st maximum in both kidneys after about 14.5 s; maximum concentration values varied between 17 and 125 µmol/mL in the aorta and between 4 and 15 µmol/mL in the kidneys). The glomerular filtration rate was calculated from the concentration changes of the contrast medium using a modified Rutland-Patlak plot technique. The GFR was 12.7 ± 2.9 mL/min m2 BS for the left kidney and 12.0 ± 2.2 mL/min/m2 BS for the right kidney. The mean values of the coefficient of determination of the regression lines were averagely 0.91 ± 0.08.
The propagation of contrast medium bolus could be depicted well. The contrast medium proceeded in a similar manner for every individual dog. Additionally, the evaluation of the single kidney function of the individual dogs is possible with this method. A standardized examination procedure would be recommended in order to minimize influencing parameters.
In companion animal medicine, the importance of canine nephropathies should not be underestimated. The prevalence of chronic kidney disease (CKD) have been assessed to be up to 3.74% [1,2,3,4]. The progressive course of this disease requires a changing treatment scheme. Therefore, periodic control examinations are recommended for an adequate treatment . Endogenous creatinine and blood urea nitrogen are used most frequently in clinical practise for evaluating the renal function but these are neither sensitive enough to reveal subclinical or border-line renal failure [6, 7], nor are they suitable for evaluating single kidney function [8, 9]. The measurement of the glomerular filtration rate (GFR) is classified as being the best single test for assessing kidney function . Despite this, this tool is seldom used in veterinary medicine because of its high cost and effort . It might also be a useful screening method for early monitoring the kidney function of dog breeds that are predisposed for nephropathies .
Inulin-clearance is reputed to be the gold standard for GFR measurement. However, this examination can only be performed under elaborate clinical conditions . Creatinine [6, 7, 12, 13] and Gadolinium-1,4,7,10-tetraazacyclododecane-1,4,7,10-tetraacetic acid (Gd-DOTA) [8, 14] are seen as eligible markers for measuring GFR in humans and rats.
As stated in numerous studies in human medicine, functional magnetic resonance imaging (fMRI) is used to assess the GFR. The main advantage is the absence of ionising radiation. This fact should not be underestimated regarding repeated evaluations of the kidney status.
Nevertheless, MRI devices are still not widely distributed in veterinary clinics and high costs for this kind of examination have to be mentioned . Additionally, limitations of renal fMRI examinations contribute to the lack of agreement on the quantification of the concentration of the contrast medium depending on signal intensity (SI) as well as a standardised protocol for the examination and analysis .
There is no consensus on a suitable model for the calculation of GFR either [17, 18]. Previous studies introduced or developed different models for GFR measurements including various amounts of compartments [18,19,20,21,22,23]. GFR was calculated in our study using a modified Rutland-Patlak plot (RPP) model. The model was used the first time by Hackstein et al.  to determine the GFR via fMRI measurements and describes a graphical solution of a simplified two-compartment model . The main advantage of this model is its simplicity because it is just a two-compartment model and no other physiological parameters are needed for calculating the GFR .
The fMRI-data were analysed to answer the following questions: (1) Is it possible to evaluate the single-kidney functions in dogs using fMRI-bolus-tracking? and (2) How much influence do different evaluation parameters have on the calculation of the modified RPP?
Eight healthy Beagle dogs (four males and four females) were included. They were kept as experimental animals at the Clinic for Small Animals at the University of Veterinary Medicine Hannover, Foundation, Germany. All procedures were approved by the animal welfare officer of the University of Veterinary Medicine Hannover, Foundation and the Lower Saxony State Office for Consumer Protection and Food Safety, Oldenburg, Germany (TV-No. 33.9-42502-04-08/1600).
The age of the dogs ranged from 4 to 11 years with a mean age of 8.5 years and a standard deviation of 2.9 years. The body weight (BW) of the dogs ranged from 14 kg to 22 kg with a mean BW of 17.5 kg and a standard deviation of 2.7 kg.
GFR measurement by the clearance of exogenously administered creatinine
To evaluate the kidneys’ health status, the dogs’ GFR was measured by determining the modified plasma-clearance of exogenously administered creatinine according to a test that has been evaluated previously [24, 25].
An exact calculated amount of 5% creatinine solution (LABOKLIN GmbH & Co.KG, 97688 Bad Kissingen, Germany) depending on the body surface (BS) of the dog was injected subcutaneously in every dog (2 g creatinine/m2 BS). BS was calculated with the help of the BW.
According to the test specifications four blood samples were taken per dog. They were sent to an external laboratory (LABOKLIN GmbH & Co.KG) and analysed there.
For analysing the fMRI measurements in vivo a functional correlation between SI and concentration of contrast medium had to be established. Therefore, a serial dilution of the contrast medium was made. 25 test tubes were filled with 0.9% NaCl and the contrast medium in concentrations from 0 to 80 mmol/L. All tubes were put into a water quench which was heated up to 38 °C and examined in MRI with the same settings as the fMRI measurements for the dogs (see Table 1—bolus track).
All MRI examinations were performed with a Philips Achieva 3 Tesla scanner.
Due to turbulence in the water, this measurement was read 4 times at intervals of 14, 106 and 127 min. The mean SIs of the different contrast medium concentrations were determined. The functional correlation between SI and concentration of contrast medium was calculated from 0 to 15 mmol/L (SI 0–1200) using the software Origin Pro® (OriginLab Corporation, Massachusetts, USA). As fit-functions a third-degree polynomial and an ascending exponential function were chosen.
MRI examination of the dogs
For preparing the anaesthesia and injecting the contrast medium all dogs were given a vein catheter either in the cephalic vein or the saphenous vein. An extension line type Heidelberger was attached to the vein catheter in order to administer the contrast medium manually during the examinations. The anaesthesia of all dogs was started with an injection of levomethadon [0.2 mg/kg intravenously (i.v.)], diazepam (1 mL/10 kg i.v.) and propofol (4–6 mg/kg i.v.). Hereafter, the dogs were intubated, and the anaesthesia was continued by inhalant anaesthesia (1–1.2% end-tidal expired isoflurane).
All dogs were in a supine position for the MRI-examinations and a body coil was used. The pre-settings for the fMRI sequence (bolus track) were chosen according to Table 1. The images of the sequence are created in a so called subtraction procedure. This means that the SIs of the third image in the sequence are taken as reference values and subtracted from all subsequent images. Every fMRI examination took about 15 min (see Additional file 1).
The contrast medium (Dotarem 0.5 mmol/mL; Querbet, 95943 RoissyCdGCedex, France) was injected as bolus at a dose of 0.1 mmol/kg (0.2 mL/kg). To insert the complete amount of contrast medium first 6 mL of 0.9% NaCl were poured into the extension line type Heidelberger. After that the contrast medium was injected and finally the extension line was rinsed with 15 mL of 0.9% NaCl.
In order to measure the SI changes of the functional MR-images, the sequences were loaded in the computer software ImageJ®. After that, manually drawn regions of interest (ROIs) were created once for all slices. The ROIs were put over both kidneys and cortices. Furthermore, a ROI was drawn over the aorta representing the vascular space. All ROIs were created within the boundaries of the tissues. Additionally, small rectangular ROIs were drawn closely to the left and right kidney as the aorta to establish a correction factor (Fig. 1). They were needed to calculate the concentrations of contrast medium in the organs.
First of all we compared the progression of SI changes in all the different ROIs of all dogs. The mean values of the SI-changes were measured until the end of MRI-examination and transferred to Excel® (Microsoft Corporation).
To convert the SIs of the different ROIs to concentrations of contrast medium in the ROIs the functional correlation of the serial dilution was used.
The slice thickness of the fMRI examinations was thicker than the volume of the relevant organs.
Therefore the complete kidney could be implied for the calculation of the GFR and not just a small part of the renal parenchym. Additionally an enlarged slice thickness leads to an improved signal to noise ratio.
Thus, SI in the voxels of the measured ROIs contained SI contributions of the organs and the residual tissues.
Therefore, the distortion caused by the residual tissue had to be corrected. In order to calculate the volume of the organs in the ROIs, the diameters of aorta, kidneys and cortices were measured in the T2-weighted images (Table 1). After that, the volume of the residual tissue was computed by subtracting the volumes of the organs. The real concentration ctissue in the tissues could be calculated according to the following formula.
cVoxel is the concentration of contrast medium in the complete voxel, VVoxel is the volume of the voxel, Vtissue is the measured volume of the tissue in the voxel, cresidual tissue is the concentration of the contrast medium in the residual space of the voxel, and Vresidual tissue is the volume of the residual tissue.
To eliminate the artefacts which were mainly caused by breathing movement a Savitzky–Golay filter  was used. The best results were found when the filter was put over 15 pictures (8.7 s).
For all dogs, the temporal changes of contrast medium concentrations were calculated and compared. Starting point for integration (t = 0 s) was the last image before a SI ascent caused by the arrival of the contrast bolus in the ROIaorta could be measured.
In order to calculate the single kidney GFR a modified Rutland-Patlak plot (RPP) was computed using the previously calculated time-dependent changes of the contrast medium concentrations in the affected organs. As stated previously  the RPP had to be modified due to the delayed propagation of the contrast medium in the renal vessels compared to the aorta. To fulfil all requirements of the RPP-model , the course of the concentrations in the kidneys was shifted by a time span (Δt). Thus, the RPP-formula was modified by Δt. The final RPP-formula sets up a straight line equation. The y-value is plotted against the x-value to calculate the gradient p.Vvas graphically. This enables the single kidney GFR to be calculated.
ckidney is the concentration of contrast medium in the kidney, Vkidney is the volume of the kidney and caorta is the concentration in the aorta. Vvas is the volume of the vascular space and p is the constant of proportionality.
The time span between the first concentration maxima in the ROIaorta and ROIkidney was taken as Δt.
Due to the heterogeneous propagation of the contrast bolus in the different dogs the starting point for integration was set to the point of time when the second maximum concentration was reached in the aorta. The time interval of the RPPs was 60 s. The starting point also marked the first pair of the x-value and y-value that was plotted in the RPPs.
In order to calculate the glomerular filtration rate of a single kidney the gradient had to be multiplied by 60 and divided by the BS in order to state the results in mL/min/m2 BS. To calculate the GFR of the plasma the results were multiplied by the factor [1 − hematocrit (hct)]. As the hct had not been measured during MRI-examination and all dogs were seen as clinically healthy, a typical hct of 0.47 was assumed according to the study of Bourgès-Abella who tried to establish reference values for Beagles, which were held under laboratory conditions .
Additionally, the influence of different sizes of ROIaorta on the calculation of the GFR was determined. As displayed in Fig. 2 two more ROIs were drawn over the aorta in order to measure the influence of size and localisation of the ROIs in the vascular space. One rectangular ROI was drawn above the bifurcation of the arteria renalis dexter (ROIa.renalis dexter) having the dimensions 3 voxels × 5 voxels; a second rectangular ROI (ROIhighest SI) of the same size was drawn over the point where the highest value of SI could be measured. A RPP was computed for these different ROIs.
For statistical analysis the computer software OriginPro® (OriginLab Corporation, Massachusetts, USA) was used. The different RPPs were compared using box-plots. Additionally, a t-test of paired samples was performed to compare the left and right kidney function.
All concentrations of the dogs’ endogenous serum-creatinine values were within the limits of the reference range as well as the results for the calculation of the “modified plasma-creatinine-clearance”. In summary, all eight tested dogs could be judged healthy with regard to the kidney function according to the reference method.
The measured values and the mean value of all four measurements can be seen in Fig. 3. The results of all four single measurements are displayed by different symbols and the mean value by a continuous line. At first SI rose with increasing concentration of contrast medium, then reached a maximum value and decreased again.
As demonstrated in Fig. 3 the correlation of SI and the concentration of the contrast medium was not linear. Although the SI exceeded 1200 for three dogs for a very short time, the fit function between SI and contrast medium was just computed in this range. The fitting in this range showed the best results for the lower concentrations and a pretty good approximation for the higher values.
The best results were found for an exponential ascent and a third degree polynomial. Comparing these two functions using the Akaike information criteria (AIC)  showed that the exponential ascent was with higher probability correct (4.8 times higher). The exponential fit curve is shown in Fig. 4 and described by the following equation:
“cGadolinium” is the concentration of Gadolinium-DOTA, “a” = − 1.933, “b” = 1.996, “x” = 0.001876 and “S” is the SI.
The time course and spatial extension of contrast medium could be seen very clearly through the whole bolus track sequence of every dog. For every ROI an unambiguous increase and decrease in SI could be measured.
With the determined equation all values for SI could be converted to definite concentrations in every ROI. As demonstrated in Fig. 5 smoothing the curves in the graphs helped to identify and interpret the extreme values of concentration of contrast medium and curve progression. The filter was set over 15 pictures all the time.
After corrections concerning the residual tissue had been performed the curve progression showed a pretty similar shape for each dog. As displayed in Fig. 6 the characteristic propagation of the contrast medium’s concentration in ROIaorta all culminated in a first maximum value after the first increase. Afterwards, concentration decreased until a characteristically minimum value. Hereafter, the concentration of contrast medium increased again until a second lower maximum and after that it decreased continuously again. The second maximum was influenced mainly by the bolus of contrast medium that ran a second time through the aorta.
A similar curve propagation could also be monitored for the ROIs of both kidneys and cortexes: in comparison to ROIaorta all a timely individually slightly delayed increase in SI could be monitored for all eight dogs.
Although the curve propagation of all dogs was of a similar nature the extreme values were reached at different point of times. Additionally, the concentration values of contrast medium varied considerably. All points of time when the extreme values in the different ROIs were reached are listed in the boxplot in Fig. 7 and the associated concentrations in Fig. 8.
The first peak of ROIaorta all occurred on average after 8.5 s, the first minimum after 23.9 s and the second maximum after 32.3 s. The maximum in all ROIs of cortexes and kidneys was reached after 14.5 s. Comparing the point of times of every single dog’s left and right first maximum resulted in a margin of 0.2 s for the ROIskidney and 0.1 s for the ROIscortex.
The comparison of concentration of contrast medium (Fig. 8) showed a wide span for ROIaorta all from 17 µmol/mL to 125 µmol/mL. The concentration in the kidneys varied from 4 µmol/mL to 16 µmol/mL and in the cortexes from 6 µmol/mL to 20 µmol/mL. The differences between the left and right side were 3.8 µmol/mL for the kidneys and 1.7 µmol/mL for the cortices.
For all eight dogs, the renal clearance of contrast medium was calculated using an RPP. The mean value of Δt, which is the time span for shifting the AIF, was 6.1 ± 0.8 s. The results of both kidneys of one dog are displayed in Fig. 9. The gradient of the trend line represents the renal clearance of the contrast medium.
A summarised presentation of all results can be seen in Fig. 10. The total GFR of our measurements amounted to an average of 24.7 ± 4.8 mL/min/m2 BS. The mean value of the left kidney was 12.7 ± 2.9 mL/min/m2 BS and on the right side 12.0 ± 2.2 mL/min/m2 BS. R2-values of the regression lines were on average 0.91 ± 0.08.
Another aspect of analysing the results of the RPPs was the choice of the size and localisation of ROIaorta. The localisations of the different ROIsaorta can be seen in Fig. 1. The results are displayed in box-plots in Fig. 11. The localisation of ROIaorta for determining the area under the curve (AUC) had a great influence on the calculated results of the RPP.
The mean values ranged from 24.7 ± 4.8 mL/min/m2 BS (ROIaorta all) to 26.2 ± 17.6 mL/min/m2 BS (ROIa.renalis dexter) to 30.3 ± 18.3 mL/min/m2BS (ROIhighest SI). The size of the ROI had an effect on the results, too. With an increasing size of the ROIs the spread of the results was minimised.
For all eight dogs, a timely similar but individually diverse progression of contrast medium could be monitored according to the different ROIs: The time gap of 7.0 s for reaching the first maximum in ROIaorta and the differences for proceedings to the further extremes underline the varying propagation of the bolus in the different dogs. Obviously, if the first maximum was reached early, also the other extreme values were reached early in comparison to the other dogs.
For the calculated concentrations similar observations were found with an even higher dispersion of the results. On the one hand, these grave fluctuations of the bolus propagation in the different dogs could be explained by the fact that the bolus was administered manually. This kind of administering is never as exact as an automatic administration. On the other hand, variations in blood pressure and depth of anaesthesia should be considered as influencing factors as well as the body weight.
As demonstrated in Figs. 7 and 8 hardly any differences could be monitored for the comparisons between the left and right kidney of each dog concerning extreme values of time or concentrations. When comparing the ROIscortex to ROIskidney also just marginal differences were detected. This aspect emphasises that the comparison of single kidney function with this method should give a clue on lower kidney function in comparison to the other side.
As demonstrated in previous studies there is no final consensus concerning the evaluation interval, which should be used for calculating the slope of the RPP [19, 29]. In this study, the ideal evaluation interval in the RPPs should be identified. In order to fulfil all requirements of the RPP-model, the delayed increase in the contrast medium in the kidneys in comparison to the aorta has to be considered.
Due to the different propagation of the contrast medium bolus in the dogs, the identification of the ideal evaluation interval for the RPPs was difficult. For about the first 30 s after the arrival of the contrast medium in the kidneys the perfusion of the renal vessels has the greatest influence on the measurements of the SIs. To rule out this influence, the best starting point for integration was identified at that point in time when the second maximum of the Gd-DOTA concentration in the aorta could be measured. For all dogs, an ascending regression line could be measured after this point in time in the RPP. This regression line reveals the clearance of the contrast medium.
The end of the evaluation interval was set after 60 s. After about 70 s for some dogs a decrease in data values in the RPPs occurred. This might be caused by the elimination of the contrast medium into the urinary bladder. At this point in time not all pre-conditions of the RPP-model would be fulfilled.
The evaluation interval of the RPP was set 60 s after the second maximum in the aorta as more than 100 data points were included for the calculation and the R2 values (R2 = 0.91 ± 0.08) of the regression lines of the RPPs indicate a good model fit for so many data points. This time interval fits well to the results described for 99mTc-DTPA in scintigraphy. They found the best time interval for integration to be between 30 and 120 s .
As presented in Fig. 10 the comparison between left and right kidney function resulted in pretty similar outcomes. For all other dogs the difference never exceeded 5 mL/min/m2 BS and was statistically insignificant (p = 0.65). This corresponded to our expected results of the healthy dogs.
As described above a so-called aortic input function (AIF) is needed to determine the concentration of contrast medium in the vascular space. Besides the susceptibility of blood vessels to artefacts in MRI measurements  there is a great impact of the choice of ROIaorta on the computed concentrations . Even in human medicine there is no standard process for calculating the AIF to enable an interindividual comparison .
The demonstrated results in Fig. 11 emphasise these aspects in our study, too. This leads to a need for standardised choice of ROIs for comparisons between single patients and different compartment models.
In this study, the results obtaining the lowest standard deviation were found when ROIaorta was selected as large as possible.
Comparisons between the GFR results of our study and the creatinine clearance method are difficult to be performed. In our study, the results from measuring the GFR via fMRI had much lower values than those from measuring the creatinine clearance.
A main point is the impact of the anaesthesia on the GFR measurements via fMRI-examinations. Lower results for GFR-measurement have been detected for patients under anaesthesia than without [33, 34]. Additionally, the results for the GFR are dependent on the method used for the measurement and the contrast medium.
Another difference between the creatinine clearance method and the fMRI measurements was the period of time in which GFR was calculated. The time interval for estimating GFR via exogenous plasma creatinine clearance was at least 4 h, whereas the period of time for measuring GFR via MRI bolus track was 60 s. Additionally, the single kidney function could only be measured by the fMRI measurements and not by a blood sampling strategy.
One challenge of the MRI bolus track evaluation was to eliminate the artefacts caused mainly by breathing movement. Two different approaches could be selected. It is possible to reduce breathing movements during examination by deep anaesthesia and manual breathing triggering. Another approach would be to eliminate these artefacts during post-editing of the images. Instead of manual corrections , a Savitzky–Golay filter  was used. As demonstrated in Fig. 5 the use of the filter showed good approximations concerning curve propagation and extreme values. This filter presented itself as a simple and non-time-consuming method for eliminating these artefacts without losing information about curve propagation or extreme values. The use of the filter also simplified the identification of the “point after aortic rise” because for some dogs breathing movement occurred at the same time as the propagation of the contrast medium bolus started.
The slice thickness of 45 mm had some advantages and disadvantages. A positive aspect of this approach was that changes of SI could be measured for the whole kidney. Furthermore, both kidneys and the aorta were completely captured in one slice so that the calculation was not limited to a small proportion of the renal parenchyma  and the contrast medium remained completely in the ROIs during the time interval of analysis . Further possible error sources might occur due to the usage of the additional ROIs that had to be established for calculations of the correction factors. As the slice thickness has a wider range than the volume of the organs the residual tissue has to be subtracted. This might lead to some falsifications because of the possible localisations of these ROIs especially for ROIaorta.
General disadvantages of this method are the great influence of the size and localisation of the ROIs on the calculations and the possible nephrotoxicity of the contrast medium  especially if repeated examinations are scheduled. A further general weakness is the lack of validated single kidney GFR reference values for dogs and the non-existence of published data about a clinically relevant change in fMRI-GFR measurements . This complicated the interpretation of the calculated results extremely.
The choice of the sectional plane affects the results, too. If a coronal sectional plane is chosen for MRI measurement the options of ROI selection are more restricted due to anatomical facts. In our opinion the definition of ROI selections is one of the key issues to establish a clinical routine measurement method as there are a lot of differences according to breed in the dog population. This might cause some challenges in defining standardised ROIs.
Another advantage is the short duration of this method compared to different blood sample methods.
The propagation of the contrast medium bolus could be depicted well. The propagation of the contrast bolus proceeded in a similar manner for every individual dog, too. Additionally, the comparison of the single kidney function of the individual dogs is possible with this method.
A standardised examination procedure (anaesthesia-protocol, administering the contrast medium automatically, localisations and size of the ROIs and determining the time interval for the integration of the RPPs) would be recommended in order to minimise influencing parameters.
As the ideal vision of the kidney examination would be a procedure in which the kidneys could be judged morphologically and functionally in one examination, MRI examinations of the kidneys seem to be a promising tool to solve this problem.
Sosnar M, Kohout P, Ruzicka M, Vrbasova L. Retrospective study of renal failure in dogs and cats admitted to University of Veterinary and Pharmaceutical Sciences Brno during 1999–2001. Acta Vet Brno. 2003;72:593–8.
O’Neill DG, Elliott J, Church DB, McGreevy PD, Thomson PC, Brodbelt DC. Chronic kidney disease in dogs in UK veterinary practices: prevalence, risk factors, and survival. J Vet Intern Med. 2013;27:814–21.
Macdougall DF, Cook T, Steward AP, Cattell V. Canine chronic renal disease: prevalence and types of glomerulonephritis in the dog. Kidney Int. 1986;29:1144–51.
Ettinger SJ, Feldman EC. Textbook of veterinary internal medicine diseases of the dog and the cat. 7th ed. St. Louis: Elsevier Saunders; 2010.
Polzin DJ. Chronic kidney disease in small animals. Vet Clin N Am-Small. 2011;41:15–30.
Von Hendy-Willson VE, Pressler BM. An overview of glomerular filtration rate testing in dogs and cats. Vet J. 2011;188:156–65.
Watson ADJ, Lefebvre HP, Concordet D, Laroute V, Ferré J-P, Braun J-P, et al. Plasma exogenous creatinine clearance test in dogs: comparison with other methods and proposed limited sampling strategy. J Vet Intern Med. 2002;16:22–33.
Zeng MY, Cheng YS, Zhao BH. Measurement of single-kidney glomerular filtration function from magnetic resonance perfusion renography. Eur J Radiol. 2015;84:1419–23.
Nikken JJ, Krestin GP. MRI of the kidney—state of the art. Eur Radiol. 2007;17:2780–93.
Finco DR, Brown SA, Vaden SL, Ferguson DC. Relationship between plasma creatinine concentration and glomerular filtration rate in dogs. J Vet Pharmacol Ther. 1995;18:418–21.
Gleadhill A, Peters AM, Michell AR. A simple method for measuring glomerular filtration rate in dogs. Res Vet Sci. 1995;59:118–23.
Braun JP, Lefebvre HP, Watson ADJ. Creatinine in the dog: a review. Vet Clin Path. 2003;32:162–79.
Cortadellas O, Fernandez del Palacio MJ, Talavera J, Bayon A. Glomerular filtration rate in dogs with leishmaniasis and chronic kidney disease. J Vet Intern Med. 2008;22:293–300.
Baumann D, Rudin M. Quantitative assessment of rat kidney function by measuring the clearance of the contrast agent Gd(DOTA) using dynamic MRI. Magn Reson Imaging. 2000;18:587–95.
Leyendecker JR, Clingan MJ. Magnetic resonance urography update—are we there yet? Semin Ultrasound CT MRI. 2009;30:246–57.
Chandarana H, Lee VS. Renal functional MRI: are we ready for clinical application? Am J Roentgenol. 2009;192:1550–7.
Zhang JL, Rusinek H, Chandarana H, Lee VS. Functional MRI of the kidneys. J Magn Reson Imaging. 2013;37:282–93.
Buckley DL, Shurrab AE, Cheung CM, Jones AP, Mamtora H, Kalra PA. Measurement of single kidney function using dynamic contrast-enhanced MRI: comparison of two models in human subjects. J Magn Reson Imaging. 2006;24:1117–23.
Hackstein N, Heckrodt J, Rau WS. Measurement of single-kidney glomerular filtration rate using a contrast-enhanced dynamic gradient-echo sequence and the Rutland-Patlak plot technique. J Magn Reson Imaging. 2003;18:714–25.
Annet L, Hermoye L, Peeters F, Jamar F, Dehoux J-P, Van Beers BE. Glomerular filtration rate: assessment with dynamic contrast-enhanced MRI and a cortical-compartment model in the rabbit kidney. J Magn Reson Imaging. 2004;20:843–9.
Sourbron SP, Michaely HJ, Reiser MF, Schoenberg SO. MRI-measurement of perfusion and glomerular filtration in the human kidney with a separable compartment model. Invest Radiol. 2008;43:40–8.
Lee VS, Rusinek H, Bokacheva L, Huang AJ, Oesingmann N, Chen Q, et al. Renal function measurements from MR renography and a simplified multicompartmental model. Am J Physiol Renal Physiol. 2007;292:F1548–59.
Zhang JL, Rusinek H, Bokacheva L, Lerman LO, Chen Q, Prince C, et al. Functional assessment of the kidney from magnetic resonance and computed tomography renography: impulse retention approach to a multicompartment model. Magn Reson Med. 2008;59:278–88.
Höchel J, Finnah A, Velde K, Hartmann H. Epidemiologie—Bewertung einer modifizierten Plasma-Clearance mit exogenem Kreatinin als ein für die Kleintierpraxis geeignetes Verfahren der renalen Funktionsdiagnostik. Berl Muench Tieraerztl Wochenschr. 2004. p. 420–7.
Hartmann H, Mohr S, Thure S, Hochel J. Routine use of a renal function test for the quantitative determination of glomerular filtration rate (GFR) including the determination of the cut-off value for azotemia in the dog. Wien Tierarztl Monatsschr. 2006;93:226–34.
Savitzky A, Golay MJE. Smoothing and differentiation of data by simplified least squares procedures. Anal Chem. 1964;36:1627–39.
Bourges-Abella NH, Gury TD, Geffre A, Concordet D, Thibault-Duprey KC, Dauchy A, et al. Reference intervals, intraindividual and interindividual variability, and reference change values for hematologic variables in laboratory beagles. J Am Assoc Lab Anim Sci. 2015;54:17–24.
Akaike H. Information theory and an extension of the maximum likelihood principle. In: Petrov BN, editors. Proceedings of the second international symposium on information theory; Budapest. Akademiai Kiado. 1973. p. 267–81.
Bokacheva L, Rusinek H, Zhang JL, Chen Q, Lee VS. Estimates of glomerular filtration rate from MR renography and tracer kinetic models. J Magn Reson Imaging. 2009;29:371–82.
Kampa N, Wennstrom U, Lord P, Twardock R, Maripuu E, Eksell P, et al. Effect of region of interest selection and uptake measurement on glomerular filtration rate measured by Tc-99m-DTPA scintigraphy in dogs. Vet Radiol Ultrasound. 2002;43:383–91.
Cutajar M, Mendichovszky IA, Tofts PS, Gordon I. The importance of AIF ROI selection in DCE-MRI renography: reproducibility and variability of renal perfusion and filtration. Eur J Radiol. 2010;74:E155–61.
Mendichovszky IA, Cutajar M, Gordon I. Reproducibility of the aortic input function (AIF) derived from dynamic contrast-enhanced magnetic resonance imaging (DCE-MRI) of the kidneys in a volunteer study. Eur J Radiol. 2009;71:576–81.
Boscan P, Pypendop B, Siao K, Francey T, Dowers K, Cowgill L, et al. Fluid balance, glomerular filtration rate, and urine output in dogs anesthetized for an orthopedic surgical procedure. Am J Vet Res. 2010;71:501–7.
Bovée KC, Joyce T. Clinical evaluation of glomerular function: 24-hour creatinine clearance in dogs. J Am Vet Med Assoc. 1979;174:488–91.
Daghini E, Juillard L, Haas JA, Krier JD, Romero JC, Lerman LO. Comparison of mathematic models for assessment of glomerular filtration rate with electron-beam CT in pigs. Radiology. 2007;242:417–24.
Pollard RE, Puchalski SM, Pascoe PJ. Hemodynamic and serum biochemical alterations associated with intravenous administration of three types of contrast media in anesthetized in dogs. Am J Vet Res. 2008;69:1268–73.
Mendichovszky I, Pedersen M, Frøkiær J, Dissing T, Grenier N, Anderson P, et al. How accurate is dynamic contrast-enhanced MRI in the assessment of renal glomerular filtration rate? A critical appraisal. J Magn Reson Imaging. 2008;27:925–31.
JNM performed measurements, analyzed data, drafted and wrote the manuscript. ML contributed to the study design, data analysis, interpretation and helped with editing. ACB and PD conducted the MRI-examinations. HS and PW contributed to the study design. PW helped with editing. All authors read and approved the final manuscript.
The authors wish to thank Mrs. F. Sherwood-Brock for proofreading the manuscript.
The authors declare that they have no competing interests.
Availability of data and materials
Data are available from the corresponding author upon request.
Consent for publication
Ethics approval and consent to participate
All procedures were approved by the animal welfare officer of the University of Veterinary Medicine Hannover, Foundation and the Lower Saxony State Office for Consumer Protection and Food Safety, Oldenburg, Germany (reference number TV-No. 33.9-42502-04-08/1600).
This work was funded by Deutsche Forschungsgemeinschaft (DFG, German Research Foundation), LU 1366/3-1.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Mehl, J., Lüpke, M., Brenner, A. et al. Measurement of single kidney glomerular filtration rate in dogs using dynamic contrast-enhanced magnetic resonance imaging and the Rutland-Patlak plot technique. Acta Vet Scand 60, 72 (2018). https://doi.org/10.1186/s13028-018-0423-3
- Dynamic contrast-enhance MRI
- Glomerular filtration rate
- Renal function
- Rutland-Patlak plot